- Automated cell growth and culture studies using BMG LABTECH microplate readers equipped with ACU
- Neisseria meningitidis used to test Atmospheric Control Unit as it requires optimally 5% CO2 to grow and thrive
- Improved growth rates observed in dilute cultures using the ACU as compared to using a CO2 controlled incubator
Table of contents
It is well known that all organisms need a certain, but not necessarily a similar, level of carbon dioxide for growth and reproduction. The period during which this level is being increased corresponds to the lag phase as the organism is unable to divide until the critical concentration of CO2 is reached.
The meningococcus has been considered one of the most fastidious microorganisms with respect to growth requirements and it has long been recognized that most strains of Neisseria meningitidis require or benefit by, a concentration of CO2 greater than atmospheric. Therefore, this organism is especially suitable for the study of CO2 effect.
In this study, a strain of Neisseria meningitidis was used to assess the efficiency of a BMG LABTECH multi-mode plate reader coupled with an Atmospheric Control Unit (ACU) to deliver 5% CO2. This was done by comparing growth of Neisseria meningitidis in the BMG LABTECH reader with a carbon dioxide incubator (set to deliver 5% CO2) and in an incubator at atmospheric CO2.
Materials & Methods
- BMG LABTECH microplate reader with ACU
- Lucy 1, Anthos, reader without atmospheric control
Neisseria meningitidis C751 was grown on Brain-Heart Infusion (BHI) agar supplemented with 10% foetal bovine serum (FBS) at 37°C and in 5% CO2 overnight. The following day, several colonies were resuspended in 10ml BHI broth supplemented with 10% FBS. This was serially diluted to 10-6 and 200 μl samples of each dilution were dispensed in triplicate into a sterile 96-well microplate. Absorbance readings were taken of each well every hour (ABS filter 405nm) in the BMG LABTECH plate reader with ACU (set at 37°C, 5% CO2) over a 24h time period.
Identical microplates were placed in a 37°C, 5% CO2 incubator and a 37°C incubator with no supplemental CO2, and absorbance readings (at 405nm) were taken at 0, 1, 2, 3, 4, 6, 8 and 24h using an Anthos Lucy 1 microplate reader.
Once absorbance readings had been collected for all treatments, the blank media control (values not shown) was used to adjust values to represent increase in OD. Percentage bacterial growth was then calculated using the maximum OD achieved for each treatment as 100%. Data was taken from triplicate wells on two separate occasions to give six data sets for analysis. Microsoft Excel was used to plot graphs and Minitab 13 was used to carry out tests of statistical significance.
Results & Discussion
From the data shown in Figures 1-3 it is evident that there is a clear correlation between starting inocula and growth rate for all three treatments.
At the higher starting inocula there appears to be little difference in bacterial growth between treatments. This is most likely due to there being a sufficiently high number of organisms present to produce a critical level of CO2, therefore enabling initiation of growth without the need for an external CO2 source.
However, differences in growth rate of N.meningitidis C751 as a result of CO2 effect are more apparent in the most dilute cultures. For example, when grown at 37°C in atmospheric CO2 (figure 1), the most dilute culture (1 in 1,000,000 dilution) exhibits a prolonged lag period due to the inability of the small inocula to reach the critical level of CO2 to initiate growth. This lag phase is appreciably shorter when the same culture is grown in either a carbon dioxide incubator (figure 2) or the BMG LABTECH plate reader with ACU achieving 29% and 72% bacterial growth respectively after 24 hours incubation.
By comparing the data collected using a carbon dioxide incubator and the new BMG LABTECH’s ACU (figure 2) it can be seen that there is less difference between growth rates of the serial dilutions using the BMG LABTECH plate reader. This is most likely due to a constant temperature and level of CO2 being maintained throughout the duration of the experiment. Whereas, when using a CO2 incubator the samples have to be transferred to a microplate reader to measure absorbance resulting in both CO2 level and temperature fluctuations which ultimately result in decreased growth rate of N.meningitidis.
The results shown in Table 1 illustrate that this increased growth rate in the BMG LABTECH plate reader compared with a carbon dioxide incubator is statistically significant for all dilutions.
The results in Table 1 also reveal a significant difference in bacterial growth between the BMG LABTECH plate reader at atmospheric CO2 for all except one dilution (1 in 10: p=0.052). However, the results also reveal that there is no significant difference in bacterial growth between a CO2 incubator and atmospheric CO2 which is most likely attributable to the temperature and CO2 level fluctuations described earlier.
Table 1: Comparing growth of a serially diluted culture of Neisseria meningitidis C751 over a 24 hour period when grown at 37°C in a carbon dioxide incubator delivering 5%, in a BMG LABTECH plate reader with ACU delivering 5% CO2 and with no supplemental CO2 (atmospheric incubator). Minitab 13 was used to carry out t-tests and results were considered significant (+) or not significant (-) at the 95% confidence interval.
|Comparison||CO2 Incubator vs BMG LABTECH||BMG LABTECH vs atmospheric incubator||CO2 Incubator vs atmospheric incubator|
|Neat||0.035 (+)||0.040 (+)||0.908 (-)|
|1 in 10||0.036 (+)||0.052 (+)||0.927 (+)|
|1 in 100||0.010 (+)||0.017 (+)||0.960 (-)|
|1 in 1,000||0.014 (+)||0.034 (+)||0.789 (+)|
|1 in 10,000||0.014 (+)||0.025 (+)||0.860 (+)|
|1 in 100,000||0.005 (+)||0.002 (+)||0.675 (-)|
|1 in 1,000,000||0.003 (+)||0.000 (+)||0.738 (-)|
In conclusion, by utilizing the CO2 dependency of a strain of Neisseria meningitidis it has been demonstrated that the BMG LABTECH plate reader with ACU is able to both achieve and sustain a level of CO2 required for growth of such a fastidious organism. This study has also highlighted the advantage of this automated system over a carbon dioxide incubator. For example, conditions are more stable and the fact that it is less labour intensive means that a greater amount of data can be generated. This system is not only ideal for experiments using CO2 dependent organisms such as N.meningitidis but may also prove very useful in cell culture studies.