- The SPARCL assay provides a platform for simpliﬁed development of immunoassays
- The reading at time of injection capability of the FLUOstar® Omega allows the assessment of the entire SPARCL signal
Table of contents
Biomarkers are used in diverse scientiﬁc ﬁelds to evaluate normal and pathogenic processes or the response to therapy. Here we look at the biomarker interleukin 8 (IL-8) which is often associated with inﬂammation and has been shown to be increased by oxidative stress. Pharmacokinetic (PK) assays are used to measure the concentration of a drug that has been administered to a human or animal. PK assay data are used to evaluate elimination proﬁles of the drug and are important in drug safety and toxicology studies.
In this application note we will show how the performance of the FLUOstar Omega microplate reader enables the detection of SPARCL (Spatial Proximity Analyte Reagent Capture Luminescence) assays. Examples of biomarker detection and pharmacokinetic assays are presented.
SPARCL detection technology was used for rapid immunoassay development. This technology is a proximity dependent and homogeneous method where an antibody/antigen interaction leads to proximity of a chemiluminescent substrate (acridan, bound to antibody 1)) and an oxidative enzyme (horseradish peroxidase (HRP, bound to antibody 2)). Introduction of a trigger solution containing H2O2 generates a ﬂash of light proportional to the amount of analyte present (Figure 1).
Materials & Methods
- SPARCL kits (Lumigen)
- FLUOstar Omega microplate reader (BMG LABTECH)
- 96-well, white, ﬂat bottom microplates from Greiner
- Other reagents were obtained from commercial sources
A typical Lumigen SPARCL assay workﬂow is shown below.
For each experiment one antibody was labeled with acridan according to SPARCL kit instructions while a second HRP-labeled antibody was purchased.
A 1:2 dilution series of IL-8 was prepared from a starting concentration of 1000 pg/mL. Dilution of controls and the dilution series were made in cell culture media or PBS containing 0.1% BSA and 10% human plasma.
A standard curve was generated using a dilution series of human IgG in neat (undiluted) rat serum. The highest point in the standard curve was 1000 ng/mL. Quality Control samples (QC) were made by spiking human IgG drug into neat rat matrix. QC concentration values were interpolated from the standard curve. The interpolated QC sample values were used to determine accuracy, precision and total error.
For both experiments sample plating, antibody addition, incubation and addition of background reducing agent were performed according to instructions.1 The plates were then read on the FLUOstar Omega plate reader with the following settings which include injection of the trigger solution.
FLUOstar Omega instrument settings
|Measurement type:||Luminescent (well mode)|
|No. of intervals:||50|
|Interval time:||0.02 s|
|Measurement interval time:||0.02 s|
|Injection volume:||75 µl|
|Pump speed:||300 µl/s|
Results & Discussion
The FLUOstar Omega features reading at time of injection which allows for the assessment of signal produced in the SPARCL reaction upon addition of the trigger solution (Figure 2).
The data produced by the SPARCL IL-8 biomarker experiment was evaluated using results based on calculating the area under the curve. As seen in Figure 3 this method is suitable for assessing the SPARCL reaction providing a reliable ﬁt curve.
Similarly SPARCL PK assays using IgG as a “drug” were performed. Figure 4 shows the representative curve produced from a dilution series of human IgG.
Assay precision and reproducibility were assessed for the PK assay using replicates of QC samples tested in 2 different SPARCL assay test runs as shown below (Table 1).
|Number of measurements||6||6||6|
|Total Error (%RE +%CV)||8.6||6.6||12.5|
*%RE = Percent relative error; **CV = coefficient of variation
Lumigen’s SPARCL assay enables simpliﬁed immuno-assay development when paired with the reading at time of injection capabilities of BMG LABTECH micro-plate readers such as the FLUOstar Omega. Both the IL-8 and the PK SPARCL assays show acceptable accuracy, precision and total error. SPARCL may be formatted for many different assay types including high throughput screening applications.
1. For more information about SPARCL contact Lumigen at SPARCL@beckman.com or visit www.lumigen.com/detection-technologies/sparcl
©2015 Lumigen, Inc. All rights reserved. Used with permission. Lumigen and SPARCL are trademarks of Lumigen, Inc. and are registered in the USPTO. Lumigen is a Beckman Coulter company.